THE DESIGN AND CONSTRUCTION OF THE HEARING AID DEVICE
Hearing aid device is a small electronic gadget that is fit in or behind the ear to improve one’s hearing and consequently communication ability.This research work involves the design and development of a hearing aid device withpre-amplifier; an acoustic signal picked-up using a condenser microphone. TDA 2822M IC is configured to produce an audio amplification which is converted to audio signal through a headphone. Designed equations were employed to calculate the physical parameters of the circuit. After the design, the circuit was constructed and tested on 4 people with partial hearing problem. The result showed that there was a significant improvement in the hearing ability of all the patients tested. The final test showed that the device could prove very useful for people with partial hearing problems. However, the device had no impact on persons suffering from complete deafness. For further improvement, it is recommended that a wireless hearing aid device should be designed and constructed to reduce the weight. The casing of the hearing aid should be made up of more portable, qualitative and lighter plastic material so as to reduce the overall weight and size. During the design, more care should be taken so as to avoid feedback signal between the microphone and the headphone which may result in noisy signal.
1.1 BACKGROUND TO THE STUDY
Hearing is one of the five senses along with vision, taste, smell and touch. The ear serves asa receiver of incoming sounds. Although the ear receives sound and is the sense organ for hearing, it is the brain and central nervous system that interpret the sound. The human auditory system can generally hear sounds within the frequency range of 20 Hz and 20 kHz but the frequency range between 100Hz and 6 kHz contains most of the information of human voice (Kinsler, Frey, Coppers, and Sanders, 1982). Although the sensation of hearing requires an intact and functioning auditory portion of the central nervous system as well as working ear, human deafness (extreme insensitivity to sound) and hearing loss most commonly occur because of damages of the ear, rather than the central auditory system.
Hearing aid device is a small electronic gadget that is fit in or behind the ear to improve one’s hearing and consequently communication ability (Yusuf, 2015). While some people are born with hearing problem some others develop it as they grow. This problem can occur as a result of disease, aging, and injury from noise or intake of certain medicines (N.I.D.C.D, 2010). Hearing problems could be that of complete deafness or partially impaired type. Hearing problem could occur after a person learned to talk (post lingual) or those with trilingual deafness that is, deafness that occurs before a person learns to talk. Deafness, whatever the degree or course, is generally a source of worry and frustration to the patient concerned as it affects almost all aspects of one’s life.
Hearing loss is usually reserved for people who have relative insensitivity to sound in the speech Frequency range. A hearing loss can happen when any part of the auditory system is not working in the usual way (Oyler, R., Oyler, A., and Matkin,, 1988). The design and constructed hearing aid device can improve the quality of sound for people with hearing loss. The heart of the circuit is an integrated circuit which amplifies all the signals picked by the condenser microphone and converts them back into sound and sent to the ear. It is a battery-powered electronic device that can amplify sound for people with hearing loss. The amplification function of the hearing aid is made possible with several components; a microphone (input transducer) which picks up sound an amplifier which increases the volume or the sound, a receiver (output transducer) which changes the electrical signal back into sound and sends it to the ear (Lowenberg,1976).
DOWNLOAD THE FULL PROJECT
How do I get this complete project on THE DESIGN AND CONSTRUCTION OF THE HEARING AID DEVICE?
Simply click on the Download button above and follow the procedure stated.
I have a fresh topic that is not on your website. How do I go about it?
How fast can I get this complete project on THE DESIGN AND CONSTRUCTION OF THE HEARING AID DEVICE?
Within 15 minutes if you want this exact project topic without adjustment
Is it a complete research project or just materials?
It is a Complete Research Project i.e Chapters 1-5, Abstract, Table of Contents, Full References, Questionnaires / Secondary Data
What if I want to change the case study for THE DESIGN AND CONSTRUCTION OF THE HEARING AID DEVICE, What do i do?
Chat with Our Instant Help Desk Now: +234 814 010 7220 and you will be responded to immediately
How will I get my complete project?
Your Complete Project Material will be sent to your Email Address in Ms Word document format
Can I get my Complete Project through WhatsApp?
Yes! We can send your Complete Research Project to your WhatsApp Number
What if my Project Supervisor made some changes to a topic i picked from your website?
Call Our Instant Help Desk Now: +234 814 010 7220 and you will be responded to immediately
Do you assist students with Assignment and Project Proposal?
Yes! Call Our Instant Help Desk Now: +234 814 010 7220 and you will be responded to immediately
What if i do not have any project topic idea at all?
Smiles! We've Got You Covered. Chat with us on WhatsApp Now to Get Instant Help: +234 814 010 7220
How can i trust this site?
We are well aware of fraudulent activities that have been happening on the internet. It is regrettable, but hopefully declining. However, we wish to reinstate to our esteemed clients that we are genuine and duly registered with the Corporate Affairs Commission as "PRIMEDGE TECHNOLOGY". This site runs on Secure Sockets Layer (SSL), therefore all transactions on this site are HIGHLY secure and safe!